"mado " <firstname.lastname@example.org> wrote in message news:email@example.com... > any help with daq ???
You didn't say in your original message in this thread what happened. All you said was " the range should be changed to +10 to -10 but that didn't happened" and that's not detailed enough to know what help you're looking to receive. [Did you get a range of -5 to 5? 0 to 10? -20 to 20? Did your hardware start smoking or explode? Did it give you unexpected results in the correct range?]
The code you posted is probably also going to be impossible to run without that specific piece of hardware, which limits how many CSSM readers are going to be able to help. You should contact Technical Support (using the instructions in my signature) and state exactly what happened and what you expected and work with them to determine the cause of the behavior you're seeing and a workaround or solution.